Virus eliminator

ABSTRACT

The enclosed application titled Virus Eliminator is described physically and functionally. This permits the elimination of viruses in the blood stream. The scope of the invention has many ramifications of being used in a clinical practice of medicine, including private physicians, hospitals and clinics.

CROSS-REFERENCE TO RELATED APPLICATIONS

[0001] Not applicable

BACKGROUND—FIELD OF INVENTION

[0002] This invention relates to human blood, when exposed toultraviolet light, can be purged of many unwanted cells that causediseases.

BACKGROUND—DESCRIPTION OF PRIOR ART

[0003] Photo chemotherapy with psoralen-containing plan extracts wasemployed in Egypt and India in 1500 B.C. for the treatment of vitiligo.El Mofty at the University of Cairo first used a purified psoralen forthe treatment of vitiligo in 1947. In 1974, Parrish et al. reportedsuccessful treatment of severe psoriasis with 8-methoxypsoralen (P) andUVA, and coined the acronym PUVA. PUVA has been approved for thetreatment of vitiligo and psoriasis. Its widespread use with extensivefollow-up has provided comprehensive data on toxicity and efficacy.

[0004] Currently a process known has helix developed by CerusCorporation has apparently developed this apparatus which usesultraviolet light and renders killer viruses bacteria harmless. Thisprocess will be used in conjunction with the Red Cross and blood centersto screen pathogens and viruses in donated blood for blood replacementin the human body. This method has been developed to treat largequantities of blood where the end result will be a product that is freeof contamination.

[0005] Presently a number of companies are in different stages ofdeveloping kidney dialysis machines and/or virus inactivation devices.These machines will be used in hospitals and blood centers. They areexpensive to operate. They require approximately six hours to radiatethe blood of an individual which is complicated procedure, timeconsuming and expensive.

[0006] This particular medical device would be portable and inexpensiveto own or to purchase and/or operate. It is designed to render atreatment rather than perfect a product.

SUMMARY

[0007] This device is capable of eliminating a number of viruses fromhuman blood through the exposure of ultraviolet light as described inthe diagrams.

Objects and Advantages

[0008] Accordingly, in addition to the advantages already described, thefollowing summary is included:

[0009] (a) to provide a treatment on an individual basis for individualswho have various contaminations including viruses.

[0010] (b) Provides a very rapid and economical method.

[0011] (c) It is mobile and easily operated.

[0012] (d) Relatively inexpensive to operate.

[0013] (e) It takes relative a shorter period of time for adecontamination of blood.

[0014] (f) It offers the individual recipient an opportunity to reducehis state of contagion.

[0015] Still further objects and advantages will become apparent from aconsideration of the ensuing description in the claim and the drawings.

DRAWING FIGURES

[0016]FIG. 1, No. 100 shows lateral view of entire unit includingelectrical outlet, quartz tube ultraviolet light.

[0017]FIG. 2, No. 100 shows opposite view from above.

[0018]FIG. 3, No. 100 shows end view including electrical outlet quartztubing and ultraviolet light.

[0019]FIG. 4, No. 100 shows opposite end.

[0020]FIG. 5 shows quartz tubing enlarged with serrations cut into tube.

[0021]FIG. 6 shows outlet on magnified scale of quartz tubing.

[0022]FIG. 7 shows quartz tube on lateral view with enlarged serrations.

[0023]FIG. 8 shows lateral view from the opposite side.

[0024]FIG. 9 shows sagital section of quartz tubing showing insidediameter of tube.

[0025]FIG. 10, No. 100 shows quartz tubing in function position,electrical outlet, electrical connections for ultraviolet tube and base,which holds tube.

[0026]FIG. 11 shows holder for ultraviolet light.

[0027]FIG. 12 shows holder for ultraviolet light including screws forsupporting base.

[0028]FIG. 13 shows supporting stand for ultraviolet light.

[0029]FIG. 14 shows lateral view of supporting stand for quartz tube.

[0030]FIG. 15 shows lateral view of supporting base for ultravioletlight. Two views: one end view and one oblique.

REFERENCE NUMERALS IN DRAWINGS

[0031] A list of Reference Numerals is included

DESCRIPTION OF FIGURES

[0032] Every reference numeral in the specification is on the drawings.

Method of Operation

[0033] Helical Quartz Blood Treatment Coil

[0034] Utilizing “indents” impressed into quartz tubing angled 45° toaxis of blood flow provides full exposure of human blood cells toultraviolet light due to turbulence caused by the indents.

[0035] Human blood, when exposed to ultraviolet light, can be purged ofmany unwanted cells that cause diseases. Therefore it is the purpose ofthis invention to provide maximum exposure of ultraviolet light to humanblood cells.

[0036] Quartz is a mandatory material for the tubing, as ultravioletlight will pass through quartz unimpeded.

[0037] When blood is passed through a straight quartz tube, some cellsare protected by the shadow of other cells and are not eradiated by theultraviolet light.

[0038] Straight tubing of a given length equates to exposure time to theultraviolet light but due to that length, the tubing is long andcumbersome for an easy-to-use device that is convenient.

[0039] Bending the quartz tubing into a helical (or spiral) shapereduces the size of the apparatus without reducing the physical lengthof the tubing.

[0040] Drawings depict a three-coil unit but any number of coils can beadded to vary the exposure time to eradiate cells that otherwise wouldnot be neutralized in a short time span.

[0041] Elimination of the “shadow” effect within the tubing is achievedby the inclusion of “indents” in the coil. These indents are impressedinto the quartz tubing while the quartz tubing is in a soft state duringthe forming of the coil shape.

[0042] These indents are impressed at a forty-five degree (45°) anglerelative to the direction of flow. These indents cause the blood tospiral within the quartz tubing providing a “mixing” effect allowingfull exposure of all cells to the ultraviolet light.

Component Identification for Each Figure

[0043]FIG. 1 (Front view)

[0044] Number

[0045]100 . . . Blood treatment unit complete.

[0046]12 . . . Quartz tubing

[0047]14 . . . Indent (6 ea, in each coil) 3 coils.

[0048]16 . . . Ultraviolet lamp.

[0049]18 . . . Spring clip lamp (16) support.

[0050]20 . . . Soft rubber support for quartz coil (12).

[0051]22 . . . Rubber cap protecting lamp (16) electrical contacts (24).

[0052]24 . . . Not shown in FIG. 1 (See FIG. 10).

[0053]26 . . . Electric supply wire for lamp (16).

[0054]28 . . . Plastic base (or other material) to house electriccomponents for lamp (16).

[0055]30 . . . Electrical plug to power lamp (16).

[0056]FIG. 2 (Top view)

[0057] All numbers are same as FIG. 1 except item (24) & (26) not shownin this view.

[0058]FIG. 3 (Right end view)

[0059] All numbers are same as FIG. 1 except item (24) not shown in thisview.

[0060]FIG. 3 (Right end view)

[0061] All numbers are same as FIG. 1 except item (24) not shown in thisview.

[0062]FIG. 4 (Left end view)

[0063] All numbers are same as FIG. 1 except item (24) not shown in thisfigure.

[0064]FIG. 5 (Right side view)

[0065]12 . . . Quartz Coil.

[0066]14 . . . Indents.

[0067]34 . . . Medical tubing to and from source.

[0068]FIG. 6 Cross section of end (in/out)

[0069]32 . . . End of Quartz tubing inside diameter reduced to sharp endto preclude restriction of fluid when entering or exiting coil.

[0070]34 . . . Medical tubing.

[0071]FIG. 7 (Side view of indent)

[0072]12 . . . Section of Quartz tubing.

[0073]14 . . . Indent in Quartz tubing.

[0074]FIG. 8 (Top view of indent)

[0075]12 . . . Section of Quartz tubing.

[0076]14 . . . Indent in Quartz tubing (top view).

[0077]FIG. 9 (End view of Quartz tubing)

[0078]12 . . . Cross section of Quartz tubing to show inside of tubingat juncture of indent.

[0079]14 . . . Indent from inside Quartz tubing.

[0080]FIG. 10 (Exploded view of 100)

[0081] All numbers same as FIG. 1 but adding item (24).

[0082]24 . . . Connection wire to ultraviolet lamp.

[0083]FIG. 11 (Side view)

[0084]18 . . . Lamp spring clip support.

[0085]FIG. 12 (Top view)

[0086]18 . . . Lamp spring clip support.

[0087]FIG. 13 (End view)

[0088]18 . . . Lamp spring clip support.

[0089]FIG. 14 (End view)

[0090]20 . . . Rubber mount pad for Quartz tubing.

[0091]FIG. 15 (Side view)

[0092]20 . . . Rubber mount pad for Quartz tubing.

[0093]FIG. 16 (Isometric view)

[0094]20 . . . Rubber mount pad for Quartz tubing.

1. The entire part consists of a piece of quartz tubing bent on itselfin a circular fashion having an entrance and an exit for the flow ofblood and containing a number of small indentations on the internallumen of the tubing. This structure and design assures the greatestamount of exposure of the blood within the tube to the effect of ultaviolet light